Potts Law Firm filed a ...
Water & Wastes Digest examines two types of samplers and their associated myths
As most readers of Water & Wastes Digest are aware, a
fluid sampler is essentially a smart pump that takes a liquid sample from a
source and deposits it into one or several containers for inspection. Fluid can
either be pulled up the intake hose, or pushed from the source. Two types of
samplers are typically used, vacuum/compressor and peristaltic.
Both peristaltic and vacuum samplers pull fluid from the
source. As each pump removes air from the intake hose, fluid from the source
flows in to equalize the pressure difference. The faster the pump can remove
air from the tube, the faster the liquid can be transported through the system.
Ideally, physics dictates that a liquid can be pulled a
maximum of 32 ft vertically, however, friction and other factors reduce that
In applications where it is necessary to lift a sample near
or greater than 32 ft, pumps can be placed near the source liquid to push
samples through the intake tubing or a collection trough. This method has no
physical limitations, save the pressure the pump is able to generate and the
strength of the tubing. A pressure of roughly 15-psi is needed for every 33 ft
of vertical climb. Losses for friction and altitude are also present.
Vacuum samplers use a vacuum/compressor to provide the
vacuum and pressure that move the liquid. A chamber is used to hold and measure
the liquid sample before deposit. The compressor forces air into this chamber,
purging it and the intake hose of fluid, then draws liquid into the chamber
until it is full.
Pressure is then applied to the chamber, expelling the
excess liquid above the preset sample volume. The sample is then deposited into
the sample bottle. This process reduces cross contamination by diluting any
liquid remaining in the intake line.
There are several manufacturers of vacuum samplers. One of
these manufacturers, Manning Environmental Inc., claims that one of the several
advantages of the vacuum system is its precise volume repeatability. The method
of completely collecting the sample volume in the vacuum chamber reduces the
amount of variation between samples (within 5 ml) regardless of factors that
would typically interfere with a sample such as head height, bubbles or other
Another distinct advantage of vacuum samplers is the high
transport velocity. Falling well within the EPA recommended range, vacuum
samplers can achieve transport velocities in excess of 5.5 ft per second, which
enable them to perform a wider range of functions. Much longer draws are
possible with vacuum samplers, making it feasible to keep personnel clear of
Vacuum systems also have no consumable parts that need
regular replacement, and because the fluid does not travel through the pump
unit, large particles or solids will not damage the sampler.
The compressor in a vacuum sampler will also generate up to
a 45-psi purge. This is useful when dealing with high fiber, rag or other
material that can clog the intake tube. This purge uses a high cubic feet per
minute rate causing more liquid to be cleared from the wetted parts and
reducing cross contamination.
Finally, vacuum samplers lend themselves to variations in
the diameter of the through path. The increased rate of flow enables the use of
a 5?8-in. through path, as well as the standard 3?8-in. Greater
diameter through paths are possible, but not commonly used.
In contrast to vacuum samplers, peristaltic samplers use two
or more rollers to push against a flexible tube creating a pillow of fluid or
As the rollers move, the pillow is pushed toward one end of
the tube. The pillow exits the pump and the next pillow is then pushed through.
Peristaltic units are well suited for toxic applications.
Using a Teflon intake hose or other approved material is usually the only
modification a peristaltic sampler requires to sample toxic fluids.
Peristaltic units are less sensitive to set up than vacuum
units, so proper intake hose placement and accurate calibration are generally
the only critical steps to accurate sampling.
Peristaltic samplers also have fewer parts, and are more
enclosed than vacuum units, making them fairly rugged and easy to operate and
maintain. Due to the type and number of parts required for peristaltic
samplers, they also tend to be priced less than vacuum samplers.
Several common misconceptions exist about the performance
and functionality of vacuum samplers.
First, vacuum samplers cross contaminate more than
In truth, results are equal between the two types. The majority of wetted surface area in
a sampler is in the intake hose, which is cleared better with the high
pressure/high volume purge of a vacuum.
A USGS test in Madison, Wis., tested this with a distilled
water blank in both vacuum and peristaltic samplers and the conclusion was both
units were equal in cross-contamination.
Another myth involves higher velocity and stronger purge, as
found in vacuum samplers, will disturb the source bed load more so than a
slower, lighter purge, and stronger suction will grab heavier particles into
the sample more than a less powerful suction.
This is untrue on both accounts. Stronger purge and suction
does not change the level of disruption of the bed load or cause a higher
particulate count than slow purge and sample units. Placing the intake hose
directly on the bed load would cause higher levels of disruption and
particulate intake from the vacuum unit, but only marginally more than a
sampler with lower purge and suction strength. In both situations, intake hose
placement is vital to accurate sampling.
A final myth is associated with strong suction concentrating
solids. Strong suction does not in any way concentrate solids. The physics of
moving a less dense particulate, or bio-solid, through a more dense substance
such as water, requires a force to act independently on the individual particulate.
A hose drawing fluid acts on the whole and material of all densities are moved
together. Any vacuum or peristaltic sampler creating stronger suction will
actually give an equal or better representation of the source liquid. This is
supported by the EPA report on samplers recommending higher transport velocity,
or stronger suction. It is further supported by the USGS test in Madison, Wis.,
which tested and found that stronger suction did not concentrate solids.
When choosing a sampler, correct analysis of your
application is vital. Peristaltic samplers are convenient for short draw
applications, toxic applications and for those who need a less-expensive
sampling solution, while vacuum samplers offer a higher level of performance
that can be used in more demanding applications.
WWD would like to recognize Larry Kolbert, sales manager, at
Manning Environmental Inc. for his consultation and contribution to this
For further information on samplers, contact Manning at